Verification and calibration of handheld mechanical or digital force gauge made by the japanees manufacturer Imada Co. Ltd., traceable to PTB/DKD/ILAC standards. The comprehensive calibration certificate contents the calculation of uncertainty, the confirmation and advise for calibration intervals according the chosen accredited calibration procedures according VDI/VDE directive 2624 sheet 2.1 or DAkkS-DKD-R 3-3 (B).
The calibration for tensile and compression force (pull and push direction) is performed on an automated calibration stand with direct staggered mass application for forces up to 1000N. The lowest resolution is 0,2N for force gauges with 2N measuring range. Force gauges with nominal measuring range higher 1kN are calibrated on a stand with transfer standards up to 250 kN.
Please consult the site "Calibration Laboratory for Force" and the part no. CAL-100 or CAL-210 in order to check which measuring ranges of the calibration stands are covered by the accreditation in relation to the nominal force range of your gauge.